We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Country:||Central African Republic|
|Published (Last):||9 February 2008|
|PDF File Size:||9.50 Mb|
|ePub File Size:||5.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
The heparin pump should be monitored to verify the intended infusion during treatment.
Fresenius Medical Care 2008K Operator’s Manual
A minor blood leak has been frewenius by the blood leak detector. Blood Pump Calibration Alarm Turn machine power off and back on.
Diasafe Filter if Applicable Assembly, installation, adjustment, or repair is to be performed only by persons authorized by the facility medical director or by Fresenius USA. Replacing The 9-volt Battery Fail Get Neg Tmp Labels And Mechanical Condition It can be set to deliver heparin in a bolus dose and at a consistent rate during the treatment.
BVM will no longer pass information to the K monitor. Temperature Control Calibration method 4 Control The control panel is located at the top third of the K and contains the Panel touch screen and panel keys used in controlling the treatment.
The dialysate delivery machine can be used with a variety of blood-tubing configurations. Temperature Control Calibration method 2 Fresenius Medical Care Freseniks The K can be configured so that this testing is mandatory after power up providing that the force test option is selected.
FRESENIUS MEDICAL CARE K OPERATOR’S MANUAL Pdf Download.
Start key a second time and the pump segment will be released from the pump head. K Conductivity High, 5 Replace the concentrates if it appears that the fluid is being continued pulled in, but the conductivity is still high. Dialysate should already be verified for absence of disinfectant, verification fresenijs prescription, conductivity, and pH should also be confirmed. The vertical dotted line indicates the scheduled end of treatment. Section 25 – Troubleshooting Power Distribution Each component contributes to the task of transmitting the blood from the patient, through the dialyzer and back to the patient.
Alarms And Troubleshooting Section 28 — Heat Exchanger Leaking Externally Section 21 — Hydraulic Leaks This data box displays in hours and minutes the amount of time the patient has been on dialysis.
UF rate will be maintained even with low blood 2008m speeds. Other Options There are other options on these screens that may not be specifically described here. Appendix Manal Concentrate Manuap The table consists of four columns: The heparin administration system is now ready for patient treatment.
New Fresenius k semi & annual pm manual – Topic
Figure 34 — Heparin Delivery Screen on page If there is no blood alarm present, pressing and holding this key for one full second will re-center the arterial and venous alarm limits.
Dialogue Box displays the current time, patient blood pressure and pulse and the time taken. Manufacturer’s Emc Declaration Calibrate If Replaced Turn the machine OFF.
Deaeration And Loading Pressure Calibration Temperature Fredenius Calibration Hold for one second to turn the power off and If blood is sensed, the machine will power down with an audible alarm. Air machine cannot be reset. If the alarm is not cleared, refer to a qualified technician.
It is now time to connect the patient to the K via the blood tubing and begin the dialysis treatment. Conventions A button refers to pressure-sensitive area located in the treatment screens that is used to set treatment parameters.
I am just looking at the machine to use as a study guide for School. During heat disinfection, the RO unit, if used, can be disconnected fresfnius the K once the water temperature reaches 80 C and the recirculation cycle starts.