View and Download Mitsubishi Electric FR-E instruction manual online. FR- E Inverter pdf Inverter Mitsubishi Electric Freqrol-A Instruction Manual. Part No: FR-ESK-EC, Brand: Mitsubishi, Model: E Series – kW V Download the Mitsubishi E Manual and Installation Guide from the links. changes both to this manual and to the specifications and design of the hardware at any . struction Manual of the frequency inverter FR-E This Installation.
|Published (Last):||13 October 2011|
|PDF File Size:||6.96 Mb|
|ePub File Size:||1.18 Mb|
|Price:||Free* [*Free Regsitration Required]|
MITSUBISHI ELECTRIC FR-E INSTRUCTION MANUAL Pdf Download.
For the way of changing the operation mode, refer to page The inverter station numbers may be set between H00 and H1F stations 0 and 31 in hexadecimal.
Daily inspection must be performed to prevent any fault from occurring due to adverse influence by the operating environment, such as temperature, humidity, dust, dirt and vibration, changes in the parts with time, service life, and other factors. Set necessary parameter values according to the load and operating specifications. Output to indicate that the process value Upper limit output signal exceeded the upper limit value. Reverse Rotation Prevention Selection pr. Page – Slip compensation Pr.
Page Page – Monitoring reference Pr.
With the factory settings, the inverter is designed to perform simple variable-speed operation. Press the key, to choose frsqrol MODE parameter setting mode. Ground fault detection is made only immediately after the start signal is input to the inverter.
Enter text from picture: Relay surface on contacts. Enter the 4 e500 20mADC process value signal Process value input from the detector. Page Page – Meter frequency meter calibration Pr For preventive maintenance, the parts must be changed periodically.
Page Page – Applied motor Pr. Always frreqrol the instructions before using the equipment 6. Their characteristics are adversely affected by ripple current, etc.
Eliminate the cause of the error preceding this error Corrective action indication. Fr-esc to sc-ene; fr-essc to sc-ene 38 pages. Handle the inverter properly in accordance with the information in each section as well as the precautions and instructions of this manual to use it correctly. By pressing ‘print’ button you will print only current page. Incorrect handling might cause an unexpected fault. Installation And Wiring 2. Combined Operation Mode 2 3.
Appropriate peripheral w500 must be selected according to the capacity.
Instruction Manual Fr-e500 Mitsubishi FREQROL Inverters
The power supply cable size of the AC reactor range motor indicated assumes that its length is 20m Stop operation when changing their parameter settings.
E50 action Reduce the load weight. By repeating step 2 below during motor run, speed can be varied. Feeqrol operation Read Operation When the inverter is operated in an ordinary, air-conditioned environment, mannual the capacitors about every 5 years. Incorrect connection will cause inverter damage or an accident. Incorrect handling may cause the inverter to operate incorrectly, causing its life to be reduced considerably, or at the worst, the inverter to be damaged.
The PU disconnection detection function judges that the PU is disconnected when it is removed from the inverter for more than 1 second. Page Page – Offline auto tuning function Pr. Though the parameter values of other than the motor constant Pr. Use FFT to measure the output voltage accurately. Parameter Function Details 4. The frequency setting signal is set using the optional control panel or parameter unit. Page – Remote setting function maunal Pr.
Reverse rotation For example 1 Inverter status b3: When the carrier frequency exceeds 5kHz, do not use the instrument because overcurrent losses occurring in the metallic parts inside the instrument will increase and may lead to burnout.
Selection of Soft-PWM will make it unoffending. Motor need not be connected. These settings differ in starting method.
Page of Go. Page 20 – Removal and reinstallation of the access Page – Stop selection Pr. Set them according to the operation specifications, load, etc. Page – Troubleshooting Page – Speed greatly differs from the setting Page – Operation mode is not changed properly Page – Precautions for Freqroll and Inspecti Page Page – User group selection Pr.
Push either A or B in the direction of arrows, and using the other end as a support, pull the front cover toward 5e00 to remove. Make calibration using Pr.